Greater than 10,000 defibrillators within the UK could also be affected by a fault, which suggests they won’t work in an emergency.
The fault impacts LIFEPAK 1000 units, the Medicines and Healthcare merchandise Regulatory Company (MHRA) mentioned.
The producer, Physio-Management, has despatched a security alert to these with the units involved, urging them to verify the battery connection.
The defibrillators are meant to be used by anybody in public locations.
They’re present in faculties, airports, leisure centres, hospitals, clinics and ambulances.
Physio-Management mentioned it had obtained 34 stories of incidents the place clients had tried to make use of the defibrillator and it had shut down unexpectedly throughout remedy.
The MHRA mentioned the fault was attributable to an intermittent connection between the battery and system contacts due to put on and corrosion.
This might fail to ship an electrical shock to resuscitate a affected person, it mentioned.
There are 10,068 LIFEPAK 1000 units within the UK.
The MHRA mentioned any customers who had but to obtain the security alert ought to contact Physio-Management immediately.
Pointers for beneficial use say LIFEPAK 1000 batteries needs to be eliminated and reinstalled each week.
What’s a defibrillator?
- It is a system that offers a excessive vitality electrical shock to the guts via the chest wall to somebody in cardiac arrest
- This excessive vitality shock known as defibrillation, and it is an important step in saving a life
- If somebody has had a cardiac arrest, name 999 and begin chest compressions or CPR
- Then discover out if there’s a defibrillator close by
John Wilkinson, MHRA director of medical units, mentioned: “These units ship life-saving remedy and it’s important they function accurately when wanted in an emergency.
“People who find themselves accountable for them ought to perform the checks beneficial by the producer.
“You probably have any questions, please contact Physio-Management on 0808 258 0094.”
In October, the identical firm issued a security warning over two different defibrillator fashions.